USER MANUAL 60 _____________________________________________________________________________________________________________________ Order of operations: Press MENU key. When StAt is displayed press T key. The following options are displayed: 0 kg - StA Prn – monitoring and printout of statistical data, StA oFF – deactivate function, StA – activate function, work with printout of chosen weighting results, StA - – activate function, work without printout, StA n – maximal samples value, Sta nM – inscribing nominal value for statistics, Sta tOL – inscribing tolerance in %, Sta tAr – automatic tare on/off StA CFG – function configuration: -Auto – Automatic work (samples are confirmed after loading the scale and indication stabilization.), -ManuAL – manual work (confirmation is made by pressing - key). out – exit from function. - Remember first to inscribe nominal weight value and tolerance (mentioned above). After that, push T key when StA o is displayed. Put on successive objects on the pan (remove after indication stabilization) in order to add them to measurements register. In order to obtain printed statistical results from measurements series press MENU key and T key when StAt is displayed and later StA Prn. After printout two options are enabled: - rESET – erasing results, - kg 0 kg Contin – continuation.