22 USER MANUAL __________________________________________________________________________________________________________________ 15.6. Function for summing recipe ingredients (rECIPE) 0.00 g MENU rECIPE T rEC oFF rEC on T o 10.01g T A B o C o When proceeding with recipe, successive ingredients (A, B, C, etc.) are weighed each time starting from zero indication, which is obtained after scale taring. 0.00g o 10.02g T If several ingredients are weighed, their sum mass can be read (despite several taring). For this purpose use rEC oFF option. 0.00g o 10.03g MENU rECIPE T rEC oFF T A+ B+ C 30.06g The function allows for separate weighing of several ingredients in one container with the possibility of reading current sum of all weighed ingredients. The function includes the following options: -rEC oFF – leave the function with the possibility of read sum mass, -rEC on – start recipe weighing, -rEC Con – continue previous recipe, -out – exit without changes. Comments: o indicator on the left side of scale display shows rECIPE function is active. SUM indicator shown when rEC oFF option is used, disappears after using →T← key.